I had a timeout issue before, however, I enabled the terminal character, and this went away. Depending on the routines that is used we have one, two or three sets of data that have to be displayed simultaneously on the same XY Graph. The program runs in a while loop and a number of data acquisition routines are accessible using a case structure. Hi, I have a project to control a motor system. I have seen situation where an old model was replaced by a new version and the advanced settings or something worked differently.
|Date Added:||17 April 2009|
|File Size:||35.99 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I don’t understand what you mean by sending 2 sets of data or 3 sets of data.
Ametek XG series power supplies – Unknown physical repeated capability selector errors. I would hp294a to know if it is possible to use LabVIEW to control and syncronize the operation of a conveyor belt that feeds material into a hydraulic press. Now your wire will no longer be broken in the other case. The way I have realised this is with a XY Graph indicator in the outer shell of the program and gp4294a local variable of it in the respective cases in that I write the sets of data using ‘build array’.
Normally is carriage return set to slash code display I have formatted the modbus command and write to tcp write and read the value from Labvies read.
I am working on re-setting up a labview environment in the lab I work at.
Instrument Control (GPIB, Serial, VISA, IVI) topics
Agilent A Resistance Read Timeout issues. Cannot obtain output from my controller.
It shows similar error message. The board is powered by an Agilent EA power supply and the resistance is measured by an Agilent A multimeter. Message 3 of The connection is through my computer serial interface.
Auto-suggest helps pabview quickly narrow down your search results by suggesting possible matches as you type. Message 10 of I’m not quite sure where to start with this. Most Active Hardware Boards: Most Active Software Boards: If anyone knows how to convert the code without interfering to the drivers, please, share with me: Hhp4294a anyone know which chip is on NI’s converter cable? I also checked HPA and it shows same error.
Agilent 4294 Analyzer
I did find a driver for the previous version of this impedance analyzer HP A on NI’s website, but ran into errors when using that. Message 2 of 7. Now it is always an array, with either one or two elements. I can’t remember the exact circumstances, but there was one time “Refresh” wouldn’t list all the resources.
I interprete this to mean I checked Np4294a visa interactive controller in NI-max and it works very well. I have created all the necessary subVIs for making the instrument driver. I had a timeout issue before, however, I enabled the terminal character, and this went away.
Looking for Labview driver for Agilent A (HP) – Discussion Forums – National Instruments
But still the program says that the subVi is missing. Claim or contact us about this channel. How long does it usually take to get such driver done? Message 1 of 3.
Agilent Technologies / Keysight Technologies hp4294a Analyzer
I haven’t found one. Also be forwarned that not all of those Keithley are created equal. Am I doing something wrong in my code?